According to the FDA Compliance Policy Guide No. 7133.11 (dated October 1, 1980) an x-ray system designed for and limited by its design for diagnostic purposes to only one of the following body regions is classified as "other than general purpose" for the purpose of Section 1020.31.
- Head and Neck
Some x-ray system manufacturers market other than general purpose systems with PBL even though it is not required. The question has been raised as to whether the PBL requirements of 1020.31(e)(2) would apply when PBL itself is not required.
A system in which the user depends on PBL rather than manual collimation creates an additional hazard if the PBL system does not function properly (especially if it oversizes the x-ray field). Therefore, when a radiographic system provides PBL operation, it must meet all requirements applicable to PBL systems, e.g., 1020.31(d)(1) and (2), and 1020.31(e)(1) and (2) even if PBL is not required. A system complying with the PBL requirements will be considered as complying with the requirements of 1020.31(f)(4).
Submit comments on this guidance document electronically via docket ID: FDA-2013-S-0610 - Specific Electronic Submissions Intended For FDA's Dockets Management Staff (i.e., Citizen Petitions, Draft Proposed Guidance Documents, Variances, and other administrative record submissions)
If unable to submit comments online, please mail written comments to:
Food and Drug Administration
5630 Fishers Lane, Rm 1061
Rockville, MD 20852
All comments should be identified with the title of the guidance.