- Medical Devices
Recipient NameMs. Saba Yussouf
- Invisi Smart Technologies UK LTD
112 Cumberland House
80 Scrubs Lane
- Issuing Office:
- Center for Devices and Radiological Health
Dear Ms. Yussouf:
The Food and Drug Administration has completed an evaluation of your firm’s corrective actions in response to our Warning Letter CMS #614512 dated August 24, 2021. Based on our evaluation, it appears that you have addressed the violation(s) contained in this Warning Letter. Future FDA inspections and regulatory activities will further assess the adequacy and sustainability of these corrections.
This letter does not relieve you or your firm from the responsibility of taking all necessary steps to assure sustained compliance with the Federal Food, Drug, and Cosmetic Act and its implementing regulations or with other relevant legal authority. The Agency expects you and your firm to maintain compliance and will continue to monitor your state of compliance. This letter will not preclude any future regulatory action should violations be observed during a subsequent inspection or through other means.
Cynthia J. Chang, Ph.D.
DHT 4B: Division of Infection Control and Plastic
OHT4: Office of Office of Surgical and Infection
Office of Product Evaluation and Quality
Center for Devices and Radiological Health