No. However, the facility is reminded that for each image receptor size used clinically for magnification, the system must meet the following three requirements:
- 900.12(b)(4)(iii) - system capable of operating with the grid removed from between the source and the image receptor
- 900.12(b)(6)(ii) - system provides at least one magnification from the range of 1.4 to 2.0
- 900.12(e)(5)(iii) - system resolution
If the facility performs magnification using multiple sizes of image receptors, then it must meet the requirements with each size receptor used for magnification.