On the facility’s x-ray unit, the indication of the detector size and position options is projected onto the input surface of the compression paddle. However, when the paddle is moved up and down, the indicated detector size does not change with distance. Is this an acceptable indication under 900.12(b)(10)(ii)(A)?
Yes. The size and positions indicated at the input surface should be indicative of the size and positions of the detector in the plane of the detector. Compliance could be achieved by representations permanently marked on the paddles or by a projected image that approximates the size and position of the detector.
Back to Automatic Exposure Control (AEC) Performance Testing