News & Events
FDA NEWS RELEASE
For Immediate Release: March 22, 2012
Media Inquiries: Morgan Liscinsky, 301-796-0397; email@example.com
Consumer Inquiries: 888-INFO-FDA
FDA approves LINX Reflux Management System to treat gastroesophageal reflux disease
The Food and Drug Administration today approved the LINX Reflux Management System for people diagnosed with gastroesophageal reflux disease (GERD) who continue to have chronic symptoms, despite the use of maximum medical therapy for the treatment of reflux.
GERD is a condition in which food or liquid in the stomach flows back into the esophagus. This can irritate the esophagus, causing heartburn and other symptoms.
Patients with GERD are first advised to make dietary and lifestyle changes such as losing weight, eating smaller meals and avoiding certain types of foods that may trigger symptoms. There are medications available, such as antacids and proton pump inhibitors, for those who do not respond to dietary and lifestyle measures.
Surgical treatment usually is reserved for people who have severe symptoms such as chronic GERD or a large hiatal hernia, a condition in which part of the stomach pushes upward through the diaphragm. Surgery is also an option for those who do not follow their recommended medical therapy or who want to avoid a lifetime of medical therapy.
“The LINX Reflux Management System is a sterile, single-use, surgically placed device used to treat the symptoms associated with GERD,” said Christy Foreman, director of the Office of Device Evaluation in FDA’s Center for Devices and Radiological Health. “LINX offers an option to patients and their health care providers and is an alternative to current surgical procedures.”
The LINX system is composed of a series of titanium beads, each with a magnetic core, connected together with independent titanium wires to form a ring shape. It is implanted at the lower esophageal sphincter (LES), a circular band of muscle that closes the last few centimeters of the esophagus and prevents the backward flow of stomach contents.
The force of the magnetic beads is designed to provide additional strength to keep a weak LES closed. Upon swallowing, the magnetic force between the beads is overcome by the higher pressures of swallowing forces, and the device expands to accommodate a normal swallow of food or liquid. Once the food passes though the LES, the device returns to its resting state.
The company conducted a feasibility study of 44 patients at four centers with a five-year follow-up plan. In addition, the company conducted a pivotal study of 100 patients at 14 centers with a five-year follow-up plan. Patients enrolled had GERD and chronic GERD symptoms, despite medical therapy. Results from both the feasibility and pivotal trials indicate that the benefits obtained with the LINX Reflux Management System outweigh its risks.
As a condition of approval, the company will institute a required training program to educate new users on patient selection, device implantation and post-procedural care of patients treated with LINX.
The most common adverse events experienced with the LINX included difficulty swallowing, pain when swallowing food, chest pain, vomiting, and nausea.
It is important to note that patients with LINX will no longer be able to undergo Magnetic Resonance Imaging (MRI) procedures. The magnetic beads interfere with the machine and can cause the device to be damaged and the patient to be injured.
The LINX Reflux Management System is manufactured by Torax Medical Inc. in St. Paul, Minn.
For more information:
The FDA, an agency within the U.S. Department of Health and Human Services, protects the public health by assuring the safety, effectiveness, and security of human and veterinary drugs, vaccines and other biological products for human use, and medical devices. The agency also is responsible for the safety and security of our nation’s food supply, cosmetics, dietary supplements, products that give off electronic radiation, and for regulating tobacco products.