Covidien Inc., Shiley 3.0PED Cuffless Pediatric Tracheostomy Tube
Class 1 Recall
|January 14, 2009|
Shiley 3.0PED Cuffless Pediatric Tracheostomy Tube
0806000266, 0806000858, 0806000859, 0806001328, 0807000555, 0807001444, 0807002502, 0808000387, 0808000431, 0808002285, 0810000288, 0810000289, 0810000864, 0810000865, 0810001201, 0811000228, 0811000522, 0811000523, 0811000546, 0811000547, 0811001875
This product was manufactured from July 7, 2008 through December 9, 2008 and distributed from July 24, 2008 through December 23, 2008.
|A tracheostomy is a surgical procedure to create an opening through the neck into the windpipe (trachea). A tracheostomy tube is usually placed through the windpipe to provide an airway and to provide a pathway to remove fluid from the trachea and lungs.|
Respiratory and Monitoring Solutions
6135 Gunbarrel Ave
Boulder, CO 80301
Reason for Recall:
|The company has recalled this product because of complaints received about difficulty inserting: 1) the device used to place the tracheostomy tube into the windpipe (obturator), and/or 2) a suction tube (catheter) into the tracheostomy tube. This problem may require that the tracheostomy tube be removed and replaced.|
|The company may be contacted at 1-800-635-5267.|
Covidien sent an urgent alert letter to its customers on January 14, 2009 notifying them of the problem. The company began sending a second letter to their customers on March 6, 2009 asking them to:
Class 1 recalls are the most serious type of recall and involve situations in which there is a reasonable probability that use of these products will cause serious injury or death.
Health care professionals and consumers may report adverse reactions or quality problems experienced with the use of these products to the FDA's MedWatch Adverse Event Reporting program. See the nearby link to MedWatch.