Fact Sheet for Healthcare Providers: Interpreting the D3 Ultra 2009 H1N1 Influenza A Virus ID Kit Results
The information previously contained on these webpages were authorized under the 2009 H1N1 Influenza Emergency Use Authorizations (EUAs). As of June 23, 2010, the EUAs have been terminated and this information is no longer current.
February 16, 2010
A public health emergency has been declared by the Secretary of Health and Human Services because of the outbreak of the 2009 H1N1 influenza virus, which is also referred to as swine influenza (H1N1) virus. This fact sheet will refer to the virus as 2009 H1N1 influenza virus. The Food and Drug Administration (FDA) has authorized the emergency use of the Diagnostic Hybrids D3 Ultra 2009 H1N1 Influenza A Virus ID Kit to test for the presence of 2009 H1N1 influenza viral antigens present in infected cells directly from nasal and nasopharyngeal swabs and aspirates/washes specimens or cell culture under an Emergency Use Authorization (EUA). This authorization will terminate on April 26, 2010, unless (1) it is revoked sooner, (2) the declaration of emergency is terminated sooner, or (3) the declaration of emergency is renewed for a longer term. The information in this Fact Sheet is the minimum necessary to inform you of the significant known and potential risks and benefits of the emergency use of the D3 Ultra 2009 H1N1 Influenza A Virus ID Kit.
At this time, there are no FDA-approved or cleared tests that identify the existence of the 2009 H1N1 influenza virus in clinical specimens. Previously, the FDA granted Emergency Use Authorization for other tests intended to detect 2009 H1N1 influenza virus, see http://www.fda.gov/MedicalDevices/Safety/ EmergencySituations/ucm161496.htm for more information. The current EUA authorizes the use of the Diagnostic Hybrids D3 Ultra 2009 H1N1 Influenza A Virus ID Kit to identify 2009 H1N1influenza virus infections in individuals with signs and symptoms of respiratory infection who previously tested positive for the presence of influenza A virus-infected cells by a currently available FDA-cleared direct immunofluorescence influenza A antibody device. Current information on 2009 H1N1 influenza virus, including case definitions and infection control guidelines, is available at http://www.cdc.gov/h1n1flu/. All information and guidelines, including those on testing for 2009 H1N1 influenza virus, may change as we continue to learn more about this disease. Please check CDC’s 2009 H1N1 influenza virus website regularly for the most current information.
What does it mean if the specimen tests positive for the 2009 H1N1 influenza virus?
A positive test result for 2009 H1N1 Influenza virus using the D3 Ultra 2009 H1N1 Influenza A Virus ID Kit indicates that the patient has been infected with the 2009 H1N1 influenza virus. The test does not indicate the stage of infection. Laboratory test results should always be considered in the context of clinical observations and epidemiologic data in making a final diagnosis. For guidelines on managing patients please refer to “Interim Guidance on Infection Control Measures for 2009 H1N1 Influenza in Healthcare Settings, Including Protection of Healthcare Personnel ” and “Updated Interim Recommendations for the Use of Antiviral Medications in the Treatment and Prevention of Influenza for the 2009-2010 Season ” at http://www.cdc.gov/h1n1flu/guidance/.
The D3 Ultra 2009 H1N1 Influenza A Virus ID Kit has been designed to minimize the likelihood of false positive test results. However, in the event of a false positive result, the risks to patients could include any or all of the following: recommendations to limit contact with uninfected persons (including at home or at the workplace), a prescription for antiviral medication or other therapy, the impaired ability to detect and receive appropriate medical care for the true infection causing the flu-like symptoms, or other unintended adverse effects.
What does it mean if the specimen tests negative for the 2009 H1N1 influenza virus?
Negative results do not preclude influenza virus infection and should not be used as the sole basis for treatment or other patient management decisions. The clinical features of the illness and the type and risk of exposure are the keys to making patient management decisions. A negative result from the D3 Ultra 2009 H1N1 Influenza A Virus ID test should not be interpreted as demonstrating that the patient does not have 2009 H1N1 influenza virus infection, if other aspects of the patient’s clinical presentation or recent epidemiologic exposures indicate that 2009 H1N1 influenza virus infection is likely, and diagnostic tests for other causes of acute respiratory illness are negative.
DIAGNOSTIC HYBRIDS, INC.
1055 East State Street
Athens, Ohio 45701
Healthcare providers will be contacted by Diagnostic Hybrids, Inc. in the event of any significant new findings observed during the course of the emergency use of the D3 Ultra 2009 H1N1 Influenza A Virus ID Kit.