Preliminary Public Health Notification - Fungal Keratitis Infections Related to Contact Lens Use (1st archival)
This is an archived document. It is no longer current information and has been superseded. The current document can be found online
April 10, 2006
Dear Healthcare Practitioner:
This is to inform you of a recent increase in the number of reports in the United States of a rare but serious fungal infection of the eye in soft contact lens wearers. The infection, a fungal keratitis caused by the Fusarium fungus, may cause vision loss requiring corneal transplants.
Both the Food and Drug Administration (FDA) and the Centers for Disease Control and Prevention (CDC) are investigating this situation. At this time, Bausch and Lomb has agreed to stop shipping the ReNu MoistureLoc brand contact lens solution. This Notification will be updated as more information becomes available.
In an MMWR Dispatch dated April 10, 2006, CDC stated that it received reports of 109 cases of suspected fungal keratitis in 17 different states. Although the majority of case patients have yet to be interviewed, complete data are available for 30 of them. Twenty-eight of the 30 wore soft contact lenses. Preliminary information obtained by CDC from patient interviews indicates that 26 of these patients remembered which products they used, and that all 26 reported using a Bausch & Lomb ReNu brand contact lens solution in the month prior to the onset of infection. Patients reported using a variety of different ReNu types from multiple different product lots. Five of the patients reported using other solutions in addition to the ReNu product. Nine of the patients reported wearing lenses overnight, a known risk factor for microbial keratitis. Eight required corneal transplantation. Strain typing of the organism is ongoing.
CDC and FDA are investigating these case reports. Also, investigations by CDC, state and local health departments, FDA, and manufacturers of contact lens solutions are underway to define specific behaviors or products that place contact lens wearers at increased risk for Fusarium keratitis.
Clusters of Fusarium keratitis were reported among contact lens users in Asia beginning in February 2006. At that time, Bausch & Lomb voluntarily suspended sales of its ReNu multipurpose solutions in Singapore and Hong Kong, pending their investigations, after multiple reports of Fusarium keratitis among contact lens users there.
Background on Microbial Keratitis
Microbial keratitis is a severe infection of the cornea. Risk factors for infection include trauma (generally with plant material), chronic ocular surface diseases,
immunodeficiency, and rarely, contact lens use. There are an estimated 30 million soft contact lens users in the United States; the annual incidence of microbial keratitis is estimated to be 4-21 per 10,000 soft contact lens users, depending on overnight lens use. Fungal keratitis is a condition more prevalent in warm climates; in the southernmost United States, fungal keratitis comprises up to 35% of microbial keratitis cases compared with 1% in New York. The proportion of fungal keratitis due to Fusarium spp. also varies by region, from 25-62%.
For healthcare providers:
- If a patient presents with a microbial keratitis, consider that a fungal infection may be involved and obtain a specimen for laboratory analysis.
- If appropriate, refer patients to an ophthalmologist that specializes in corneal disease for immediate treatment.
- Report cases of fungal keratitis in contact lens wearers to FDA as noted below.
For contact lens wearers:
- Wash hands with soap and water, and dry (lint-free method) before handling lenses.
- Wear and replace lenses according to the schedule prescribed by the doctor.
- Follow the specific lens cleaning and storage guidelines from the doctor and the solution manufacturer.
- Keep the contact lens case clean and replace every 3-6 months.
- Remove the lenses and consult your doctor immediately if you experience symptoms such as redness, pain, tearing, increased light sensitivity, blurry vision, discharge or swelling.
In addition, regardless of which cleaning/disinfecting solution used, wearers may want to consider performing a “rub and rinse” lens cleaning method, rather than a no rub method, in order to minimize the number of germs and reduce the chances of infection.
FDA Advice to Patients on this topic can be found online.
Reporting Adverse Events
FDA and CDC are very interested in gathering information related to fungal keratitis in contact lens users. We encourage you to report these infections to FDA. FDA will be sharing reported information with CDC. You can report directly to MedWatch, the FDA’s voluntary reporting program. You may submit reports to MedWatch by phone at 1-800-FDA-1088; by FAX at 1-800-FDA-0178; by mail to MedWatch, Food and Drug Administration, 5600 Fishers Lane, Rockville, MD 20857-9787; or online.
Please include the following information (if available) on the MedWatch reporting form:
- Contact lens solutions trade names and lot numbers.
- Contact lens type, trade name and mode of wear (extended or daily wear).
- Patient non-compliance with contact lens regimen (e.g., overnight wear in daily wear lenses, not cleaning lenses).
- Results of all cultures taken (e.g., corneal, conjunctival, contact lens, care solutions, lens case).
- Special patient characteristics, including whether the patient was immunocompromised (e.g., used topical or systemic corticosteroids or had diabetes), or had any ocular trauma, surgery, or chronic eye problem.
- Topical ocular medications used to treat the patient (including trade names and lot number if available).
Getting More Information
If you have questions about this Notification, please contact FDA's Division of of Small Manufacturers, International and Consumer Assistance (DSMICA) by e-mail at firstname.lastname@example.org or by phone at 1-800-638-2041 or 301-796-7100
FDA medical device Public Health Notifications are available on the Internet. You can also be notified through email on the day the safety notification is released by subscribing to our list server. To subscribe, visit: http://service.govdelivery.com/service/subscribe.html?code=USFDACDRH_10.
Daniel G. Schultz, MD
Center for Devices and Radiological Health
Food and Drug Administration