Guidance for Industry and FDA Staff - Class II Special Controls Guidance Document: Automated Fluorescence in situ Hybridization (FISH) Enumeration Systems
Document issued on: March 23, 2005
For questions regarding this guidance, contact Kevin Lorick at 301-796-6696 or by email at email@example.com
Written comments and suggestions may be submitted at any time for Agency consideration to the Division of Dockets Management, Food and Drug Administration, 5630 Fishers Lane, Room 1061, (HFA-305), Rockville, MD, 20852. Alternatively, electronic comments may be submitted to Regulations.gov . When submitting comments, please refer to Docket No. 2005D-0082. Comments may not be acted upon by the Agency until the document is next revised or updated.
Additional copies are available from the Internet. You may also send an e-mail request to firstname.lastname@example.org to receive an electronic copy of the guidance or send a fax request to 301-827-8149 to receive a hard copy. Please use the document number (1550) to identify the guidance you are requesting.
Table of Contents
Guidance for Industry and FDA Staff
Class II Special Controls Guidance Document: Automated Fluorescence in situ Hybridization (FISH) Enumeration Systems
This guidance document was developed as a special controls guidance to support the classification of automated fluorescence in situ hybridization (FISH) enumeration systems into class II (special controls). An automated FISH enumeration system is a device consisting of an automated scanning microscope and image analysis system designed to detect and enumerate FISH signals in interphase nuclei of formalin-fixed, paraffin-embedded human tissue specimens. The systems also contain common hardware and software platforms with customized software applications for FISH assays. Automated FISH enumeration systems are intended for in vitro diagnostic use with FISH assays as an aid in the detection, counting, and classification of cells based on recognition of cellular color, size, and shape. The use of automated systems may reduce hands-on time compared to manual enumeration of FISH assays.
This guidance is issued in conjunction with a Federal Register notice announcing the classification of (FISH) Enumeration Systems. Any firm submitting a premarket notification (510(k)) for an automated FISH enumeration system will need to address the issues covered in this special control guidance document. However, the firm need only show that its device meets the recommendations of the guidance or in some other way provides equivalent assurances of safety and effectiveness.
The firm must show that its device addresses the issues of safety and effectiveness identified in this guidance, either by meeting the recommendations of this guidance or by some other means that provides equivalent assurances of safety and effectiveness.
The Least Burdensome Approach
The issues identified in this guidance document represent those that we believe need to be addressed before your device can be marketed. In developing the guidance, we carefully considered the relevant statutory criteria for Agency decision-making. We also considered the burden that may be incurred in your attempt to comply with the statutory and regulatory criteria in the manner suggested by the guidance and in your attempt to address the issues we have identified. We believe that we have considered the least burdensome approach to resolving the issues presented in the guidance document. If, however, you believe that there is a less burdensome way to address the issues, you should follow the procedures outlined in the “A Suggested Approach to Resolving Least Burdensome Issues” document.
FDA believes that special controls, when combined with the general controls, will be sufficient to provide reasonable assurance of the safety and effectiveness of an automated FISH enumeration system. A manufacturer who intends to market a device of this generic type should (1) conform to the general controls of the Federal Food, Drug, and Cosmetic Act (the Act), including the premarket notification requirements described in 21 CFR 807 Subpart E, (2) address the specific risks to health associated with an automated FISH enumeration system identified in this guidance, and (3) obtain a substantial equivalence determination from FDA before marketing the device.
This guidance document identifies the classification regulation and product code for an automated FISH enumeration system. (Refer to Section 4 Scope.) In addition, other sections of this guidance document identify the risk to health and describe measures that, if followed by manufacturers and combined with the general controls, will generally address the risk associated with these automated FISH enumeration systems and lead to a timely premarket notification [510(k)] review and clearance. This document supplements other FDA documents regarding the specific content requirements of a premarket notification submission. You should also refer to 21 CFR 807.87 and other FDA documents on this topic, such as Premarket Notification 510(k).
As explained in "The New 510(k) Paradigm - Alternate Approaches to Demonstrating Substantial Equivalence in Premarket Notifications; Final Guidance,"1 a manufacturer may submit either a Traditional 510(k) or an Abbreviated 510(k). FDA believes an Abbreviated 510(k) provides the least burdensome means of demonstrating substantial equivalence for a new device, particularly once FDA has issued a guidance document that provides recommendations on what should be addressed in a submission for the device. Alternatively, manufacturers considering modifications to their own cleared devices may lessen the regulatory burden by submitting a Special 510(k).
An Abbreviated 510(k) submission must include the required elements identified in 21 CFR 807.87, including the proposed labeling for the device sufficient to describe the device, its intended use, and the directions for its use. In an Abbreviated 510(k), FDA may consider the contents of a summary report to be appropriate supporting data within the meaning of 21 CFR 807.87(f) or (g); therefore, we recommend that you include a summary report. The report should describe how this guidance document was used during the device development and testing and the methods or tests used. The report should also include a summary of the test data or description of the acceptance criteria applied to address the risks identified in this document, as well as any additional risks specific to your device. This section suggests information to fulfill some of the requirements of 807.87 as well as some other items that we recommend you include in an Abbreviated 510(k).
The coversheet should prominently identify the submission as an Abbreviated 510(k) and cite the title of this guidance document.
Proposed labeling should be sufficient to describe the device, its intended use, and the directions for its use. (Refer to Section 8 for specific information that you should include in the labeling for this type of device.)
We recommend that the summary report contain the following:
If it is not clear how you have addressed the risks identified by FDA or additional risks identified through your risk analysis, we may request additional information about aspects of the device’s performance characteristics. We may also request additional information if we need it to assess the adequacy of your acceptance criteria. (Under 21 CFR 807.87(l), we may request any additional information that is necessary to reach a determination regarding substantial equivalence.)
As an alternative to submitting an Abbreviated 510(k), you can submit a Traditional 510(k) that provides all of the information and data required under 21 CFR 807.87 and described in this guidance. A Traditional 510(k) should include all of your methods, data, acceptance criteria, and conclusions. Manufacturers considering modifications to their own cleared devices should consider submitting Special 510(k)s.
The general discussion above applies to any device subject to a special controls guidance document. The following is a specific discussion of how you should apply this special controls guidance document to a premarket notification for an automated FISH enumeration system.
The scope of this document is limited to the following devices as described in 21 CFR 866.4700 (product code: NTH):
21 CFR 866.4700: Automated Fluorescence in situ Hybridization (FISH) Enumeration Systems. An automated FISH enumeration system is a device that consists of an automated scanning microscope, image analysis system, and customized software applications for FISH assays. This device is intended for in vitro diagnostic use with FISH assays as an aid in the detection, counting, and classification of cells based on recognition of cellular color, size, and shape and in the detection and enumeration of FISH signals in interphase nuclei of formalin-fixed, paraffin-embedded human tissue specimens.
FDA has identified the risk to health associated with this type of device as failure of the device to perform as indicated or error in interpretation of results that may lead to improper patient management, including misdiagnosis and improper treatment. A falsely low fluorescence signal count, or false negative, could contribute to a delay in detecting the disease, disease recurrence, disease prognosis, or a false indication of response to therapy. A falsely high fluorescence signal count, or false positive, could contribute to unnecessary monitoring, inappropriate treatment decisions, or failure to treat adequately. In addition, use of assay results to adjust a treatment regimen without consideration of other clinical factors could pose a risk.
In the table below, FDA has identified the risk to health generally associated with the use of an automated FISH enumeration system addressed in this document. The measures recommended to mitigate this identified risk are described in this guidance document, as shown in the table below. You should conduct a risk analysis, prior to submitting your premarket notification, to identify any other risks specific to your device. The premarket notification should describe the risk analysis method. If you elect to use an alternative approach to address a particular risk identified in this document, or have identified risks additional to those in this document, you should provide sufficient detail to support the approach you have used to address that risk.
General Study Recommendations
We recommend that you evaluate the assay in at least three external sites. Generally, you should assess performance in the testing environment where the device will ultimately be used (i.e., central laboratory) by individuals who will use the test in clinical practice. You should initially analyze data separately to evaluate any inter-site variation and include results of the analysis in the 510(k) summary report. It may be appropriate to report pooled results from the individual sites in the package insert, if you can demonstrate that there are no significant differences in the results among sites. Before initiating a clinical study, you may wish to contact the Division of Immunology and Hematology Devices.
We recommend that you provide appropriate specifics concerning protocols so that we can interpret acceptance criteria or data summaries during the review. For example, when referring to NCCLS protocols or guidelines, we recommend that you indicate which specific aspects of the protocols or guidelines you followed. We also recommend that you include protocol specifics in labeling, as these may be necessary to aid users in interpreting information in your labeling.
We believe the software used in class II automated FISH enumeration systems may meet the definition given in these guidance documents for devices with a minor or moderate level of concern, depending on the specific analyte and impact that the software application would have on the diagnosis. Therefore, you should provide documentation for the appropriate level of concern of the device.
Specific Performance Characteristics
Because various cell selection and enumeration systems may be based on different biological selection and detection agents, and because instrumentation may differ considerably between devices, FDA recommends that, for an automated FISH enumeration system, you compare the results of your device to the reference method used for the predicate device (e.g., the cleared manual enumeration method). In addition, the assay you use to validate your automated FISH enumeration system for the 510(k) should be a legally marketed FISH assay. You may contact the Division of Immunology, Hematology and Pathology Devices for FDA input on your study plan prior to initiating comparison studies.
We recommend that you incorporate the following in your clinical evaluation study plan:
We recommend the following concerning sample size and selection in your studies:
Your 510(k) submission should include a description of your internal (i.e., manufacturer’s site) and external site protocols, and study results. You should describe how you addressed the issues concerning study plan and sample selection listed above. You should also describe the following:
The premarket notification should include labeling in sufficient detail to satisfy the requirements of 21 CFR 807.87(e). The following suggestions are aimed at assisting you in preparing labeling that satisfies the requirements of 21 CFR 807.87(e). Although final labeling is not required for 510(k) clearance, final labeling must comply with the requirements of 21 CFR 801 and 21 CFR 809.10 before a medical device is introduced into interstate commerce. Labeling recommendations in this guidance are consistent with the requirements of part 801 and section 809.10.
Directions for use
Precautions and Warnings
2 Refer to Indications for Use Form (PDF File Size: 1.03MB) for the recommended format.
3 If FDA makes a substantial equivalence determination based on acceptance criteria, the subject device should be tested and shown to meet these acceptance criteria before being introduced into interstate commerce. If the finished device does not meet the acceptance criteria and, thus, differs from the device described in the cleared 510(k), FDA recommends that submitters apply the same criteria used to assess modifications to legally marketed devices (21 CFR 807.81(a)(3)) to determine whether marketing of the finished device requires clearance of a new 510(k).
4 See Required Elements for a Declaration of Conformity to a Recognized Standard (Screening Checklist for All Premarket Notification [510(K)] Submissions).