Recall of Medical Device Notification: Sunquest Laboratory Blood Bank and Blood Donor Modules v6.0.1 and v6.0.2-Sunquest Information Systems, Inc.
DATE NOTIFICATION INITIATED:
January 23, 2008
Sunquest Laboratory Blood Bank and Blood Donor Modules v6.0.1 and v6.0.2
Distributed with Sunquest Laboratory v6.1, v6.2 and v6.3
Sunquest Information Systems, Inc.
In the Sunquest Blood Bank and Blood Donor Modules, a problem has been identified in Blood Product Testing. If the user chooses to delete and unacceptable product test result by replacing it with another unacceptable product test result, then cancels and saves, the system incorrectly retains the initial result instead of leaving the product test pending. The problem only occurs in specialized situations. Customers are asked to request the software correction via the Client support Webpage.