Device Correction of Baxter Amicus Separator-Baxter Healthcare Corp
DATE RECALL INITIATED:
March 16, 2005
PRODUCT / LOT NUMBER:
Product Codes 4R4580 and 4R4580R
All Serial Numbers
Baxter Healthcare Corporation
Baxter has redesigned the AMICUS SEPARATOR Instrument centrifuge window to withstand ammonia cleaning; however it is Baxter's recommendation that ammonia and ammonia based cleaners NOT be used to clean any part of the AMICUS SEPARATOR Instrument. Instead, a mild soap/detergent is recommended. Baxter will replace the AMICUS SEPARATOR Instrument centrifuge window on all AMICUS SEPARATOR Instruments.