DATE RECALL INITIATED:
June 14, 2005
Hemoflow Model 300
Product Code 9-3000-0xxx
Serial Numbers: B4162 thru B4167; B4225 thru B4314, and B4315 thru B4324
Applied Science, Inc
Grass Valley, CA
Hemoflow Model 300 mixer may fail to start at the beginning of a draw due to a mismatch between motor characteristics and driver circuitry. The event will trigger a "Job Error" requiring the draw to be completed manually using the SCALE mode while manually mixing the blood and anticoagulant.
Consignees have been asked to contact the manufacturer to arrange for device correction.