DATE RECALL INITIATED:
April 26, 2006
PRODUCT / LOT NUMBER
Ranger High Flow Blood / Fluid Warming Disposable Sets
Lots H2057 through H2182
Part Numbers: 24350, 24355, 24365, 24370
Eden Prairie, Minnesota
Biotest Laboratories, Inc.
High-flow disposable sets for the Ranger blood/fluid warming system may have been assembled with an inverted bubble trap. An inverted bubble trap on a Ranger high flow system may allow air to pass through the bubble trap and bypass the trap’s auto-venting mechanism, potentially permitting air to reach the patient. Consignees have been asked to contact the recalling firm to arrange for replacement product.