DATE NOTIFICATION INITIATED:
September 19, 2007
Data Innovations Instrument Manager - Software versions 7.04.0001 through 8.04.03.01
Data Innovations, Inc.
South Burlington, Vermont
Corrupt database indices in the Specimen Management database could cause the misassociation of test and test result for a specific specimen(s). Customers are requested to download the repair utility from the manufacturer's website.